Clinical Evaluation of a Portable System for Capturing Nystagmus (PSCN) as an Alternative to Videonystagmography
AbstractObjective: To develop a portable and affordable system for diagnosing pathological horizontal nystagmus (oscillatory rapid eye movement ) in remote clinics. It is aimed to be a temporary alternative to the golden standard tool which is the Video Nystagmography which is expensive and office-based equipment. Method: The team created a system composed of a hardware gadget and software application. The modified goggle with a built-in camera and microcontroller is used to record and analyze the induced nystagmus. The analyzed data is then transmitted via Bluetooth to the user portable device which displays the results. Programming PSCN, it was tested in the outpatient clinic. During the clinical evaluation of the gadget, the nystagmus was induced and recorded by the Video Nystagmography. Immediately, we repeated the experiment using the created system. Eventually, the two results were compared to set up the thresholds in the new system. After calibration, the amended algorithm of PSCN tested another ten recorded nystagmus videos to evaluate its accuracy. Results: The portable system showed high clinical reliability during the testing in comparison to golden standard Videonystagmography. In the second stage, it showed an accuracy ratio up to 80 % during testing the ten recorded videos. It also managed to differntiate the real nystagmus from the artifacts. Conclusion: This trial proofed the concept of portable system as an assistive diagnostic tool for pathological nystagmus. However, a more clinical trial is required to establish it as a portable alternative for the golden standard
Bardorf MC.2017Acquired nystagmus. Medscape. See http://medicine.medscape.com/article/1199177-overview/ (accessed 5 April 2018).
Shoup GA. Electronystagmography. Medscape 2016. http://emedicine.medscape.com/article/836028-overview (accessed 24 October 2017
Yetiser S., Ince D. Analysis of benign paroxysmal positional vertigo as a common public health problem. Annals Medical & Health Sciences Research 2015 Jan-Feb; 5 (1). https://www.ncbi.nlm.nih.gov/pmc/articles/PMC4350063/ ( accessed 20 May 2016).
Weill Cornell Medicine.Electronystagmography (ENG)/Videonystamography VNG Testing. https://ent.weill.cornell.edu/patients/clinical-specialties/conditions/electronystagmogrophy-eng-videonystagmography-vng-testing ( accessed 25 October 2017)
National Dizzy and balance center. Videonystagmography (n.d.). http://www.nationaldizzyandbalancecenter.com/services/balance-lab-testing/videonystagmography/ (accessed 10 September 2017).
Turuwhenua J, Yu TY, Mazharullah Z, Thompson B. A method for detecting optokinetic nystagmus based on the optical flow of the limbus.Vision Research, 2014, 103:75-82. https://www. ncbi.nlm.nih.gov/ PubMed/ 25151522 ( accessed 15 May 2016).
Cambridge Dictionary. https://dictionary.cambridge.org/ dictionary/english/ algorithm/ (accessed 3 April 3 , 2017)
Houcque D. Introduction to MATLAB for engineering students.2005 https://www.mccormick.northwestern.edu/documents/students/undergraduate/introduction-to-matlab.pdf ( accessed 30 March 2018).
HowStuffWorks. What is the difference between CCD and CMOS image sensors in a digital camera? 2018. https://electronics.howstuffworks.com/ camera photography/digital/question362.htm/(accessed 10 April 2018).
All opinions and reports within the articles that are published in the Gazi Medical Journal are the personal opinions of author(s). Gazi University, Editors and the publisher do not accept any responsibility for these articles. The journal is printed on acid-free paper.